Receive scope, initial inspection
Updated by tin over 8 years ago
- Status changed from New to In Progress
- % Done changed from 0 to 50
Faulty 2200uF/6.3V capacitor on x86 CPU socket area -> Replaced to new 2200uF/16V
Acquisition Board to PPC Board interface board -> tested OK (TDS7704B test pass)
Acquisition Board & PPC power board -> tested OK (TDS7704B test pass)
Acquisition Board to power Board interface board -> tested OK (TDS7704B test pass)
Power supply -> tested OK (TDS7704B test pass, same power supply).
Missing heatsink on U700 IC on ACQ Board -> suspect BGA damage?
HDD reports SMART fail. Still able to boot Win98.
CD-RW in drive contains Tekscope software ver 2.5.3
Tekscope software does not run (just shows splash screen on startup and does nothing, PPC board rools "post codes")